Deepace KC951011 vector network tester calibration parts N type KC901S open circuit breaker
KC951011 vector network tester calibration parts N type KC901S open circuit breaker
Vector network tester KC951011 type calibration device N type KC901S open circuit breaker
The main body is made of stainless steel. The interface is accurate and the parasitic parameters are small
KC951011 Reflectance standard
Nominal electric length 5.26mm.
The phase uncertainty after calibration conforms to the published target of KC901S. The open short-circuiter can operate up to 6GHz.
The load system HS products have been strictly screened, and in the range of 0~2G, the return loss is better than 42dB; in the 2G~3G , it is better than the 36dB (guarantee index); in the 3~6G is better than 30dB (this frequency band does not guarantee the target)
Typical values in 0~2GHz are better than 45dB, and 2~3GHz is better than 42dB.
This product is suitable for KC901S type vector antenna analyzer, and can also be used for calibration of other network analyzers
Design life: 500 times
Compare cycle (suggestion): every 100 times
Standard test is a valuable attachment, need cherish and care, safekeeping, regular verification. Before the calibration, the connector must be checked, cleaned to avoid damage standard. Open short-circuiter can be cleaned by water or alcohol cloth. Hold the rod and only rotate the screw sleeve, do not make the rotation of all standard when connecting. When connecting the load, you should press the golden part first, then rotate the screw sleeve to reduce the chance of the golden part rotating together
|For Vehicle Type|
|Remote Control Peripherals/Devices|
Vehicles & Remote Control Toys
|Four-wheel Drive Attributes|
|RC Parts & Accs|